|Description||OroboPOS-Service_Kit, with O2k-Service Box including all oxygen sensor service accessories for membrane mounting and service of the OroboPOS.|
|Type||OroboPOS, O2k, O2k-FluoRespirometer, Startup O2k-Respirometer|
|Link||MiPNet22.11, MiPNet19.18B POS-service|
The OroboPOS-Service Kit consists of
|Title||Description||Product id||Product image|
|O2k-Service Box||O2k-Service Box empty, contains components of the OroboPOS-Service Kit and the O2k-Assembly Kit.||26101-03|
|OroboPOS-Seal Tip||OroboPOS-Seal Tip, black butyl rubber gasket with 3 mm pore. 4/Pck, for sealing the OroboPOS (sensor head) against the O2k-Chamber. Push the wetted OroboPOS-Seal Tip over the POS-sensor head, with the pore positioned centrally, not covering any peripheral area of the gold cathode. Do not stretch the gasket across the POS-sensor head. One pack (4 seal tips) of this product is a standard component of the OroboPOS-Service Kit.||26110-01|
|OroboPOS-Membranes||OroboPOS-Membranes, FEP 25 µm; 40/Pck. Two packages of this item are standard components of the OroboPOS-Service Kit.||26200-01|
|OroboPOS-Electrolyte Powder||OroboPOS-Electrolyte Powder, KCl.
This item is a standard component of the OroboPOS-Service Kit.The powder is dissolved in 10 ml distilled water to yield a 1 M KCl solution.
|OroboPOS-Mounting Tool||OroboPOS-Mounting Tool for application of OroboPOS-Membranes. It consists of two parts, (i) the membrane guide (larger diameter) and (ii) the membrane ring holder with O-ring\Viton\6x1 mm for holding the OroboPOS-Membrane Ring during membrane application. This item is a standard component of the OroboPOS-Service Kit.||26400-02|
|OroboPOS-Polishing Cloth||OroboPOS-Polishing Cloth for cathode cleaning. Replace the OroboPOS-Polishing Cloth at intervals. A two-year interval may be considered in cases of intensive use. This item is a standard component of the OroboPOS-Service Kit.||26510-01|
|OroboPOS-Polishing Powder||OroboPOS-Polishing Powder for cathode cleaning. This item is a standard component of the OroboPOS-Service Kit.||26520-01|
|O2-Zero Powder||O2-Zero Powder, dithionite (Na2S2O4), for zero-oxygen calibration of the OroboPOS. This item is a standard component of the OroboPOS-Service Kit.||26600-01|
|Forceps for membrane application||Forceps for membrane application: for OroboPOS and ISE membrane application; do not use for tissue preparation. This item is a standard component of the OroboPOS-Service Kit and the O2k-TPP+ ISE-Module.||26700-01|
|Pipette\Plastic\1 ml ungraded||Pipette\Plastic\1 mL ungraded, for filling electrolyte into the reservoir of the OroboPOS. This item is a standard component of the OroboPOS-Service Kit.||26800-01|
* Talk:OroboPOS-Service_Kit: What is a kit?